To conduct blood pressure measurements by Korotkoff Sounds on your ALEX PCS, you’ll need to use your SmartCuff™ and your SmartScope™.
First, making sure your ALEX PCS is powered on and online, ready for use.
Start a simulation session, ALEX Healthy for example (BP 132/64 or as you set the systolic value).
Next, wrap the SmartCuff around the manikin's upper arm (either arm works, exceptions to the IV Arm), just as you would do with a real patient
Place the SmartScope on the brachial artery on the arm with the cuff
Then, inflate the cuff to around 160mmHg. If you listen through the stethoscope, you will hear nothing at this point.
Finally, slowly deflate the SmartCuff. You will start hearing the Korotkoff sounds just under 132mmHg cuff pressure (or at the set systolic value)
The Korotkoff sound stops when the cuff pressure goes under 64mmHg (or the set diastolic value)
When the cuff is fully deflated, a log entry is created saying "Blood pressure measured successfully by radial pulse palpation".
No events are created after your measurements? Click here to learn how to check the battery of your SmartCuff as a first step.
Check out how to conduct a blood pressure measurement by radial pulse palpation as well on ALEX PCS here.